You can configure the Configure Burst node with several inputs. Install Agilent Series Instrument Driver NI develops and supports thousands of instrument drivers for third party instruments to help engineers and scientists automate measurements. I have the programming guide for the generator, and I understand the idea of the command string that is fed to it, but tinkering around with this might be a waste of time for me since the IVI drivers work ok. Message 3 of 7. After doing this, the IVI drivers HPa did not turn off the output before decreasing the voltage. Finally, you can specify execution flow, reporting, database logging, and connectivity to other enterprise systems for your test system.
|Date Added:||13 June 2018|
|File Size:||39.78 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
LabVIEW driver for the 33210A
Click the Agilent 33XXX project to create a new copy. Select the PC from the SystemDesigner page. Back to Top 5. I don’t know what you are doing with it and I don’t know if you will have any delay between changes.
Keysight | Electronic Measurement Products | Agilent
TestStand, a ready-to-run test management software, is designed to help you develop, execute, and deploy automated test and validation labviw faster. I need to know when the waveform starts to be repeated, it is, when the beginning of the waveform oc The NI Instrument Driver Network houses thousands of instrument drivers for automating third-party instruments. How do I disable this?
Accepted by topic author jto2. When writing an instrument control application for a serial instrument, labbiew communicate with a COM port rather than the serial instrument connected to the COM port.
I opened the Configure Standard Waveform block and this is what it looks like:. Can the VISA driver be fixed easily?
NI develops and supports thousands of instrument drivers for third party instruments to help engineers and scientists automate measurements. Thanks for the input. Comparing the tick count to you wait seems dangerous to me. Then click the green Run arrow at the top left of the VI.
NI PXI Agi,ent Generators can produce precise waveforms including sine, square, triangle, and ramp as well as arbitrary, user-defined waveforms by using data sequences or by streaming continuously from a host or peer-to-peer instrument within the PXI system. Download Agilent Series Instrument Driver.
Automate Signal Generation From a Keysight/Agilent 33000 Function Generator With LabVIEW NXG
Customize the panel of the VI to fit your application needs. From the time you press the button, until the function takes place could be 5 sec. Thanks for the suggestion Mateus, but I am trying to automate this 33201a as best I can. Open the Learning tab by selecting the Learn to Program tile in the Welcome screen or selecting the Learning tab from the top right.
The 33xxx driver takes ms to change, and has a strange burst signal in between. Thanks last modified by shaobing Why is the output voltage twice the value I set?
Waveform Generators | Keysight Community
Drop all the indicators and controls from the Unplaced items box on the panel. I wrote a new VI based on the Configure Standard Waveform VI so that it just writes the new amplitude command, and none of the other stuff.
You can find these instrument drivers in the Instrument Driver Network. You can also troubleshoot and make manual function calls to your instrument to ensure that it is working as expected before moving into automation.
You can configure the Configure Burst node with several inputs. Message 7 of 7.